Ultrasonic flowmeter common problems (2)
5, the external clip type ultrasonic flowmeter signal is low?
This depends on the technical content of the instrument itself. After a large number of test cases on the spot, it proves that the long-term pipeline, serious scaling and large diameter, the Aitulel AFTU-2W series is better than other external-clip ultrasonic flowmeters. The signal is very fast and the signal is very stable. .
Solution: For large diameter and serious scaling, it is recommended to use a good quality external clamp type ultrasonic flowmeter. The pipe at the probe installation should be cleaned, and the air between the probe and the workpiece surface should be removed with a coupling agent or coupling piece to make the ultrasonic wave. It can be effectively introduced into the pipeline to ensure sufficient sound intensity transmission on the detection surface.
6. Occasionally, there are bubbles in the measurement medium. Does the ultrasonic flowmeter with the time difference method have any influence?
Today's Aitulel AFTU-2W series external clip type ultrasonic flowmeter has dual mode. When there is air bubble, it can be automatically transferred to Doppler mode to measure. When the bubble disappears, it will automatically turn into the time difference method.
7. The meter cannot be used under strong interference on the spot?
On-site inverter or high voltage cable field strong electromagnetic interference
Recommendation: Keep away from the inverter or high voltage cable field electromagnetic interference
8. What is the minimum diameter of the externally clamped ultrasonic flowmeter on the market? What is the highest temperature?
Answer: Now only the minimum diameter of the AFTU-2W series in the world can be measured to 6mm, the temperature can be measured at 550 °C, such as the measurement of molten salt and heat transfer oil, which is not possible with other brands of flowmeters. of.
9. How to choose a suitable ultrasonic flowmeter?
Answer: Pipe material, pipe wall thickness and pipe diameter; fluid type, contain impurities, air bubbles and full pipe; fluid temperature, flow meter type, portable or fixed online.
Intelligent GPRS MAG flow meter working principle
Since the GPRS communication is a data packet communication network based on an IP address, after the monitoring center computer logs in to the Internet network, the IP address is assigned by the Internet, and each GPRS monitoring point establishes a connection with the host through the address and communicates. Each monitoring point collects data through the data acquisition module, processes the data through the embedded processor in the GPRS terminal, encapsulates the protocol, sends it to the GPRS network, and transmits it to the user data monitoring center computer.
Battery-powered converter GPRS communication introduction
The battery-powered converter with GPRS function can directly send data to the remote control center through the GPRS wireless module GPRS-DY or GPRS-DF, hereinafter referred to as the GPRS module, and then store the collected data in the database at regular intervals. Through the system, even in an off-site away from the observation site, it is convenient to collect and read data of each monitoring point, and truly realize the functions of remote monitoring and data sharing.
Battery powered electromagnetic flowmeter features
1. Micro power consumption design, five 3.6V lithium batteries, continuous operation for 5~10 years, and easy to replace the battery.
2. The display adopts LCD large-screen liquid crystal display, which can display multiple flow parameters such as instantaneous flow rate, flow rate (pipe pressure), cumulative total amount of forward and reverse, and alarm prompt.
3. There are no movable and resistance parts in the measuring tube, no entanglement or clogging phenomenon, almost no pressure loss, and long-term reliable and continuous work.
4. Wide measurement range: The flow rate measurement range can reach 0.05m/s~10m/s.
5. High measurement accuracy: ±0.5% ~ ±1.0%.
6. Measurement stability: The measurement accuracy is not affected by changes in physical parameters such as temperature, pressure, viscosity and density of the measured medium.
7, with pulse output and GSM wireless data remote transmission.
8, with data storage function, up to 1000 groups of traffic data can be saved, and the data storage time can be arbitrarily set and data query.