Ultrasonic flowmeter common problems (2)
5, the external clip type ultrasonic flowmeter signal is low?
This depends on the technical content of the instrument itself. After a large number of test cases on the spot, it proves that the long-term pipeline, serious scaling and large diameter, the Aitulel AFTU-2W series is better than other external-clip ultrasonic flowmeters. The signal is very fast and the signal is very stable. .
Solution: For large diameter and serious scaling, it is recommended to use a good quality external clamp type ultrasonic flowmeter. The pipe at the probe installation should be cleaned, and the air between the probe and the workpiece surface should be removed with a coupling agent or coupling piece to make the ultrasonic wave. It can be effectively introduced into the pipeline to ensure sufficient sound intensity transmission on the detection surface.
6. Occasionally, there are bubbles in the measurement medium. Does the ultrasonic flowmeter with the time difference method have any influence?
Today's Aitulel AFTU-2W series external clip type ultrasonic flowmeter has dual mode. When there is air bubble, it can be automatically transferred to Doppler mode to measure. When the bubble disappears, it will automatically turn into the time difference method.
7. The meter cannot be used under strong interference on the spot?
On-site inverter or high voltage cable field strong electromagnetic interference
Recommendation: Keep away from the inverter or high voltage cable field electromagnetic interference
8. What is the minimum diameter of the externally clamped ultrasonic flowmeter on the market? What is the highest temperature?
Answer: Now only the minimum diameter of the AFTU-2W series in the world can be measured to 6mm, the temperature can be measured at 550 °C, such as the measurement of molten salt and heat transfer oil, which is not possible with other brands of flowmeters. of.
9. How to choose a suitable ultrasonic flowmeter?
Answer: Pipe material, pipe wall thickness and pipe diameter; fluid type, contain impurities, air bubbles and full pipe; fluid temperature, flow meter type, portable or fixed online.
Selection of measurement types for special industrial and commercial users
At present, there are some special users in the supply of industrial and commercial users. On the one hand, the gas points are more dispersed, and the amount of gas used in a single combustion device is small (generally 2 to 9 m3/h); on the other hand, the total amount of gas used is Very large, there are many problems with the accurate measurement of the natural gas supply of such users. This type of industrial and commercial users have multiple fuel appliances with different gas consumption. Different gas consumption at different time periods brings certain difficulties to the accurate measurement of natural gas. Under the condition of satisfying the maximum gas consumption, it cannot be measured under the condition of small flow. The problem.
1. Case 1: A large shopping and entertainment mall
The mall has 22 gas points distributed on the second, fourth and fifth floors. In order to ensure the accuracy of natural gas measurement, the gas supply system has a master meter (turbine flow meter Q650) and 22 sets of compensation functions. As a metering table, the flow meter usually has a gas consumption of about 400m3/d and a holiday of about 800m3/d. It is found through the data collection that the total number of meters and sub-meters is about 17m3 per day; in the initial 54d. The cumulative gas consumption of the master meter and the minute meter differs by 11511.80 m3.
In the case of small flow metering, the film surface ratio is the largest, the Roots flowmeter is second, and the turbine flowmeter is the smallest. The initial flow performance of the volumetric flowmeter should be superior to the velocity flowmeter. For industrial and commercial users with large total gas consumption and low heat load for each gas, the volumetric flowmeter, ie, the membrane meter and the Roots flowmeter, should be preferred to meet the flowmeter requirements of a single burner when used alone. It can also meet the measurement requirements of the maximum flow rate when all burners are used at the same time.
Not all industrial and commercial users can solve the measurement problem by selecting a volumetric flowmeter with a larger range. The volumetric ratio of the volumetric flowmeter is limited, and it is impossible to increase without limit. The selection of the flowmeter cannot be solved. This kind of user has a measurement problem. When the user only uses a single device for gas, or when operating at low flow rate, the initial flow rate and minimum flow rate of a single flow meter may not meet the minimum flow rate requirement. A single large flow meter cannot measure low gas consumption, which will result in Large supply and marketing differences.