﻿ Electromagnetic Open Channel Flow Meter

# Electromagnetic Open Channel Flow Meter

2020-09-26 11:04:14

## Electromagnetic Open Channel Flow Meter

How the turbine flow meter works
The working principle of the turbine flowmeter: the fluid flows through the sensor housing. Since the blade of the impeller has a certain angle with the flow direction, the momentum of the fluid causes the blade to have a rotational moment. After the friction torque and the fluid resistance are overcome, the blade rotates, and the rotational speed is stabilized after the torque balance. Under certain conditions, the rotational speed is proportional to the flow rate. Due to the magnetic permeability of the blade, it is in the magnetic field of the signal detector (composed of permanent magnet and coil). The rotating blade cuts the magnetic field lines and periodically changes the coil. Magnetic flux, so that the two ends of the coil induce electricity
Pulse signal, which is amplified and shaped by the amplifier to form a continuous rectangular pulse wave with a certain amplitude, which can be transmitted to the display instrument to display the instantaneous flow rate and cumulative amount of the fluid. Within a certain flow range, the pulse frequency f is proportional to the instantaneous flow rate Q of the fluid flowing through the sensor. The flow equation is: Q = 3600 × f / k
In the formula:
F——pulse frequency [Hz];
K——the meter factor of the sensor [1/m], given by the checklist. If [1/L] is used, Q=3.6×f/k
Q——the instantaneous flow rate of the fluid (under working condition) [m3/h];
3600 - conversion factor.
The meter factor of each sensor is filled in the verification certificate by the manufacturer, and the k value is set in the matching display meter to display the instantaneous flow rate and the cumulative total amount.

## Electromagnetic Open Channel Flow Meter

Selection of measurement types for special industrial and commercial users
I. Introduction
At present, there are some special users in the supply of industrial and commercial users. On the one hand, the gas points are more dispersed, and the amount of gas used in a single combustion device is small (generally 2 to 9 m3/h); on the other hand, the total amount of gas used is Very large, there are many problems with the accurate measurement of the natural gas supply of such users. This type of industrial and commercial users have multiple fuel appliances with different gas consumption. Different gas consumption at different time periods brings certain difficulties to the accurate measurement of natural gas. Under the condition of satisfying the maximum gas consumption, it cannot be measured under the condition of small flow. The problem.
1. Case 1: A large shopping and entertainment mall
The mall has 22 gas points distributed on the second, fourth and fifth floors. In order to ensure the accuracy of natural gas measurement, the gas supply system has a master meter (turbine flow meter Q650) and 22 sets of compensation functions. As a metering table, the flow meter usually has a gas consumption of about 400m3/d and a holiday of about 800m3/d. It is found through the data collection that the total number of meters and sub-meters is about 17m3 per day; in the initial 54d. The cumulative gas consumption of the master meter and the minute meter differs by 11511.80 m3.

## Electromagnetic Open Channel Flow Meter

Plug-in electromagnetic flowmeter features
Corrosion-resistant full stainless steel material, ABS full mold production probe structure is simple, firm, no moving parts, long service life, reliable measurement, strong anti-interference ability, small size, light weight, easy installation, small maintenance range, large measurement range, temperature-free measurement The influence of changes in density, pressure, viscosity, conductivity, etc., zero pressure loss can be installed and disassembled under continuous flow, which brings convenience to the user's maintenance. The manufacturing cost and installation cost are lower than the general electromagnetic flowmeter. It is especially suitable for large and medium-diameter pipeline flow measurement. It adopts advanced processing technology, solid-state packaging, vibration resistance and long life. It makes the meter have good measurement accuracy and stability. The flowmeter not only has 4-20mA current signal output, but also has 1-2000HZ frequency output digital communication output: RS485, RS232, HART, Modbus.