﻿ Electromagnetic Meter

# Electromagnetic Meter

2020-09-26 11:09:19

## Electromagnetic Meter

How the turbine flow meter works
The working principle of the turbine flowmeter: the fluid flows through the sensor housing. Since the blade of the impeller has a certain angle with the flow direction, the momentum of the fluid causes the blade to have a rotational moment. After the friction torque and the fluid resistance are overcome, the blade rotates, and the rotational speed is stabilized after the torque balance. Under certain conditions, the rotational speed is proportional to the flow rate. Due to the magnetic permeability of the blade, it is in the magnetic field of the signal detector (composed of permanent magnet and coil). The rotating blade cuts the magnetic field lines and periodically changes the coil. Magnetic flux, so that the two ends of the coil induce electricity
Pulse signal, which is amplified and shaped by the amplifier to form a continuous rectangular pulse wave with a certain amplitude, which can be transmitted to the display instrument to display the instantaneous flow rate and cumulative amount of the fluid. Within a certain flow range, the pulse frequency f is proportional to the instantaneous flow rate Q of the fluid flowing through the sensor. The flow equation is: Q = 3600 × f / k
In the formula:
F——pulse frequency [Hz];
K——the meter factor of the sensor [1/m], given by the checklist. If [1/L] is used, Q=3.6×f/k
Q——the instantaneous flow rate of the fluid (under working condition) [m3/h];
3600 - conversion factor.
The meter factor of each sensor is filled in the verification certificate by the manufacturer, and the k value is set in the matching display meter to display the instantaneous flow rate and the cumulative total amount.

## Electromagnetic Meter

Selection of measurement types for special industrial and commercial users
2. Case 2: A heavy machine tool factory
The main gas plant of the heavy-duty machine tool factory is 13 production kiln, and there is also a machine repair workshop and 63 mobile burners. The total gas load of all gas facilities is 1837m3/h. See Table 1 for details. The customer promised that all gas facilities could not be fully activated at the same time, the maximum gas consumption required was 1200m3/h, and the minimum pressure required before the furnace was 0.055MPa. The pressure regulator used in the on-site pressure regulating metering box is opened and ready, and the flow meter is in the form of a G1600 turbine table plus bypass.
When debugging the kiln, it was found that the flowmeter in the pressure regulating metering box had a stop-and-go phenomenon. If only some of the kiln was opened or only some of the kiln was used, there would be a phenomenon that the flowmeter did not measure or the measurement error was large.

## Electromagnetic Meter

Turbine flow meter applications
Turbine flow meters are widely used in the following measurement objects: petroleum, organic liquids, inorganic liquids, liquefied gases, natural gas, gas and cryogenic fluids. In the transshipment and gathering stations of foreign liquefied petroleum gas, refined oil and light crude oil, the first and last stations of large crude oil transmission pipelines use it for trade settlement. In Europe and the United States, the turbine flowmeter is the natural gas meter next to the orifice flowmeter. In the Netherlands alone, more than 2,600 gas turbine flowmeters of various sizes and pressures from 0.8 MPa to 6.5 MPa are used on the natural gas pipeline. Has become an excellent natural gas flow meter. Although the excellent metering characteristics of the turbine flow meter are favored by people, the impression is that the moving parts are short in use time, and they are inevitable in the selection. After people's unremitting efforts, it should be said that the situation has changed a lot. As the most versatile flowmeter, the turbine flowmeter has been developed into a multi-variety, full-series, multi-size mass production scale.