﻿ Digital Output Flowmeter

# Digital Output Flowmeter

2020-09-26 11:07:45

## Digital Output Flowmeter

How the turbine flow meter works
The working principle of the turbine flowmeter: the fluid flows through the sensor housing. Since the blade of the impeller has a certain angle with the flow direction, the momentum of the fluid causes the blade to have a rotational moment. After the friction torque and the fluid resistance are overcome, the blade rotates, and the rotational speed is stabilized after the torque balance. Under certain conditions, the rotational speed is proportional to the flow rate. Due to the magnetic permeability of the blade, it is in the magnetic field of the signal detector (composed of permanent magnet and coil). The rotating blade cuts the magnetic field lines and periodically changes the coil. Magnetic flux, so that the two ends of the coil induce electricity
Pulse signal, which is amplified and shaped by the amplifier to form a continuous rectangular pulse wave with a certain amplitude, which can be transmitted to the display instrument to display the instantaneous flow rate and cumulative amount of the fluid. Within a certain flow range, the pulse frequency f is proportional to the instantaneous flow rate Q of the fluid flowing through the sensor. The flow equation is: Q = 3600 × f / k
In the formula:
F——pulse frequency [Hz];
K——the meter factor of the sensor [1/m], given by the checklist. If [1/L] is used, Q=3.6×f/k
Q——the instantaneous flow rate of the fluid (under working condition) [m3/h];
3600 - conversion factor.
The meter factor of each sensor is filled in the verification certificate by the manufacturer, and the k value is set in the matching display meter to display the instantaneous flow rate and the cumulative total amount.

## Digital Output Flowmeter

Second, choose from the type of flow meter
Various flow meters, such as volumetric flowmeters, speedmeters, and differential pressuremeters, have different characteristics. Different flowmeter types have different turndown ratios. The turbine flowmeter has a turndown ratio of 1:20 to 1:30, the Roots flowmeter has a turndown ratio of 1:20 to 1:160, and the film gauge has a turndown ratio of 1: 160 (The error characteristic curves of these three kinds of flow meters are shown in Figure 1), and the appropriate flow meter should be selected according to the actual situation.
By comparing the error characteristics of the above flowmeters, it can be seen that the starting flow of the membrane meter and the Roots flowmeter is small, the initial flow of the turbine meter is relatively large, and the membrane surface exhibits a positive deviation in the small flow state. The Roots flowmeter exhibits a negative deviation. At low flow rates, the gas supply company is more willing to select a membrane meter for metering.

## Digital Output Flowmeter

In the case of small flow metering, the film surface ratio is the largest, the Roots flowmeter is second, and the turbine flowmeter is the smallest. The initial flow performance of the volumetric flowmeter should be superior to the velocity flowmeter. For industrial and commercial users with large total gas consumption and low heat load for each gas, the volumetric flowmeter, ie, the membrane meter and the Roots flowmeter, should be preferred to meet the flowmeter requirements of a single burner when used alone. It can also meet the measurement requirements of the maximum flow rate when all burners are used at the same time.
Not all industrial and commercial users can solve the measurement problem by selecting a volumetric flowmeter with a larger range. The volumetric ratio of the volumetric flowmeter is limited, and it is impossible to increase without limit. The selection of the flowmeter cannot be solved. This kind of user has a measurement problem. When the user only uses a single device for gas, or when operating at low flow rate, the initial flow rate and minimum flow rate of a single flow meter may not meet the minimum flow rate requirement. A single large flow meter cannot measure low gas consumption, which will result in Large supply and marketing differences.